This is an old revision of the document!
This page gives user-level instructions on how to run the S800 data acquisition (DAQ) system for an experiment. Presently this is run from the linux machine u6pc5 in data-U6.
The S800 Readout GUI is invoked by either clicking the icon S800 DAQ in the desktop of u6pc5 or by navigating in a Linux terminal to the directory /user/s800/converged_daq and typing
godaq. The Readout GUI window will appear with three tags labeled “main”, “SSHPipe@localhost:0”, and “SSHPipe@localhost:1”. The first one provides general information about the ReadoutGUI. The second and third tabs provide information about the data sources (from the CCUSB and VMUSB controllers). IMPORTANT NOTE: When running Readout GUI, make sure that the environment variables needed by the system are properly defined. This can be done from a Linux session connected to spdaq19 by typing:
. /usr/opt/nscldaq/xxxx/daqsetup.bash, where
xxxx is the nscldaq version (xxx = 11.0-020, on Feb 2016).
Before beginning taking data is necessary to initialize the system. This is done by clicking
Start in the ReadoutGUI. You can inspect the status of each source during their initialization by clicking in one of the tabs “SSHPipe@localhost:0” or “SSHPipe@localhost:1”. You will see a series of messages about the different initialization steps. The last message should be “Done”. During the initialization process, a fourth tab labeled “ActionFilter” may appear with information from the S800 filter. Sometimes, you may see a warning message about an old “still-running” S800 filter session being killed. That's ok.
After initializing the contollers, the ReadoutGUI window will show the
Begin button active. In addition, three additional buttons should appear at the end of the window. They can be used to start the GUI associated with the MCFD, Delay XLM, and Trigger ULM modules. Make sure that the system is completely initializing before opening those GUIs.
The figure below shows the Readout GUI window after initializing the system. After clicking
Begin, the “Event Builder” window will pop out, displaying information about the Readout session. (Note that the tab “ActionFilter” may appear if it didn't show up before, when you clicked
Start.) Data can be recorded on disk by checking out the box
Record. To end a data run, simply click
The S800 electronics includes a Mesytec CFD (MCFD), used to “filter” the detector signals going to the Scaler and Mesytec TDC. The configuration parameters of the CFD (thresholds, delays, fraction, etc.) can be remotely adjusted via a MCFD GUI developed by the NSCL DAQ group. During tuning of the S800, one typically needs to adjust thresholds only.
The MCFD GUI can be started in three different ways: 1) by clicking the button
Launch MCFD GUI in Readout GUI; 2) by clicking the icon MCFD GUI in the desktop of u6pc5; 3) by navigating in a Linux terminal and typing
$DAQBIN/MCFDControl16 –protocol usb –serialfile /dev/ttyUSB0. The environment variable DAQBIN is defined by sourcing the daqsetup.bash file in directory /usr/opt/nscldaq/xxxx/bin, where xxxx is the nscldaq version number (11.0-015 on Oct 26, 2015).
Browseand select directory /user/operations/daq/usb/Config (default)
Backto return to the main GUI
This window appears when the “Launch Gate Delay GUI” button is pressed on the Readout GUI. The delay module allows software configurable delays to be applied to each of the signals indicated in the Channel column, which then form the TDC stops. It is configurable to enable delays to be set with beam on target, as the needed delay may change depending on experimental conditions.
Some important things to remember:
The Trigger GUI appears when the “Launch ULM GUI” button is pressed on Readout GUI. The Trigger GUI is a visual display of the various Gate and Delay Generators and logic elements that make up the configurable trigger of the S800. The logic of the trigger decision is readily discerned from a visual inspection of this GUI. Setting the trigger configuration is also done using this GUI
The different signal going through the trigger scheme can be inspected in th Data-U6 oscilloscope. Simply right click on any of the wires to put that signal onto one of the four “Trigger Inspect” channels available at the patch panel of Data-U6, and connect that patch-panel cable to the oscilloscope. By inspecting the various delays, widths and overlaps the user trigger can be configured. The Trigger GUI is discussed in greater detail here.
The GUI used to display scalers rates can be open from the icon S800 Scalers in the desktop of u6pc5. Alternatively, open a terminal on u6pc5, and type ./goscalers from directory /user/s800/converged_daq/scalers.
The GUI includes two pages labeled “s800” and “ratios”. Page “s800” includes all the scaler channels; page “ratios” displays ratio values calculated between several pairs of channels. In addition, the GUI includes a panel showing the time evolution of the live time calculated from the live-to-raw trigger ratio, and the live-to-raw clock ratio. The figure below shows the page “ratios” from the scaler GUI.
A list of scaler channels can be found here.
The S800 DAQ can be run in Standalone mode (as described above), or in Slave mode. The later means that the S800 DAQ is controlled by an external DAQ (e.g. GRETINA). In this mode, the S800 ULM receives external clock and external synchronization signals from the master DAQ. Running in Slave mode requires to change the script CC0105Begin.tcl in directory /user/s800/converged_daq/Scripts. In this file, there are two variables
extclock to define if the external synchronization and clock signals are enabled (=1) or disabled (=0). Make sure that you set these variables to 1 if you want to run in Slave mode.
By default, when running in Slave mode, data are recorded in the stagearea of the Master DAQ (experiment account). It is however possible to record simultaneously data from the S800 into the S800 stagearea, using the
multilogger option (see ReadoutGUI figure above). The S800 ReadougGUI offers the possibility to record data from four different S800 ring-buffers: rawccusb, rawvmusb, s800built, and s800filter. The later is the most important since data from this ring buffer can be immediately processed by the S800 SpecTcl.
Before running an experiment with a Master system (e.g. GRETINA), it is recommendable to run the S800 DAQ in Slave mode with the s800filter multilogger enabled. This can be done following the steps:
multiloggerin the ReadoutGUI
to enable recording data from s800filter ===== Troubleshooting ===== There are several common reasons why the S800 DAQ may fails to start. The list below describes what to do to fix these problems. * Old processes still running * No data sources defined * CCUSB and/or VMUSB claimed by existing programs * Manual rebooting ==== Old processes still running ==== Sometimes, particularly if the previous S800 DAQ session ended in an uncontrolled way, there may be old processes running (CCUSB and/or VMUSB and/or S800 event builder) that will prevent Readout GUI to begin a run. Whenever this happens (typically after clickingBegin
in Readout GUI), the system sends an error message complaining about one of these processes still running. If you find yourself in this situation try: - Exit your Readout GUI properly - On a Linux session, connect to spdaq19 by typingssh -Y s800@spdaq19
(the S800 password can be found in the S800 binder that is kept in the data-U1 logbook cabinet) - Typeps aux | grep Readout
to ensure that Readout is indeed not running - Type/usr/opt/nscldaq/xxxx/bin/ringbuffer status
, where xxxx is the daq version, e.g. 11.0-020 - You will see a list of ringbuffers with information about their status (see figure below). Check that the producer value of each ringbuffer (see columnproducer
) is -1. (Don't worry about the ringbuffer s800filter, which is automatically killed when starting ReadoutGUI) - If the producer value of a ringbuffer is not -1, take note of the PID, and kill the process by typingkill -9 PID
- You should now be able to run Readout GUI and begin a run without problem ==== No data sources defined ==== When clickingBegin
in the S800 Readout GUI, you get the error message: “No data sources are running so a run cannot be started”. Very likely, the setting file .settings.tcl is missing. On a Linux session in u6pc5, typels -lisa ~/stagearea
. You should see a hidden file .settings.tcl. If no, try the following: - Go to /mnt/daqtesting/s800/exxxxlast, where exxxxlast corresponds to the experiment number of the last (successfully) run experiment - Copy the file .settings.tcl from that directory to the current stagearea. NOTE: we are assuming that the stagearea is pointing to the new experiment directory /mnt/daqtesting/s800/exxxxnew - If you cannot find that file, there is an old version that can be copied from the directory /user/s800/converged_daq - You should now be able to run ReadoutGUI without problem ==== CCUSB and/or VMUSB claimed by existing programs ==== One of the most common reasons why the S800 DAQ fails to start is because the CCUSB and/or VMUSB are claimed by existing programs. This is shown by the error messages displayed by the system. If you find yourself in this situation try the following: - Log on to S800 spdaq (spdaq19 as of October 2015) - Typeps aux | grep Readout
- Usekill -9 PID
, where PID identifies any CCUSBReadout or VMUSBReadout processes that show up - Usekill -9 PID
, where PID identifies any tclsh ReadoutShell process that might show up - Trygodaq'' to see if this works now - If this didn't help try rebooting the S800 DAQ manually (see below) ==== Manual rebooting ==== In the rare case that the RunControl GUI gets “frozen”, it is possible to reboot the whole system manually. This operation requires to turn off the VME and CAMAC crates, and the spdaq computer, all them located in the rack seating near the FP box in the S3 vault. Then, the system must be turned back on in the following order. First, CAMAC crate, second VME crate, and third spdaq.